Medical air disinfection machines usually refer to disinfection equipment for air sterilization. Common air disinfectants are mainly composed of three main components. One is the power system (the motor provides power system). The second is the filter system (HEPA high -efficiency filter, activated carbon filter). Third, the disinfection system (ultraviolet, photocatalytic). Several important factors that choose air disinfectioners are too many particles such as dust in the living environment, and haze weather that affects people’s healthy life. Second, just as the newborn baby needs a clean air environment, the physical condition of the family is relatively fragile. Third, indoor passive smoking, oil fume and other environments; fourth, there are a large amount of bacteria, viruses growing, and air needed to disinfect the spring and summer season.
How to maintain that medical air disinfectioner?
The machine is specially responsible for special personnel, and it is on standby for 24 hours to prevent frequent start -up damage. Medical air disinfection machine
Filter Management: Check the filter regularly. If the filter is found too much, it should be cleaned or replaced with water in time. During the replacement, open the filter cover, remove the filter network from the machine, replace it with a new filter, and finally reinstall the filter on the machine.
UV light management: UV radiation intensity can affect the effect of air disinfection. Therefore, at least once a month, ethanol cotton balls are applied to prevent the radiation strength of UV lamps from being affected by dust.
Management of negative oxygen ion generator: Under the effect of negative oxygen ions, it is used for a long time. A large amount of dust will be accumulated near the exhaust mouth shell of the generator, which must be cleared regularly. When cleaning, first cut off the power and wipe it with a soft dry cloth or a little medical ethanol. Note: Do not wash with water. The human body induction probe and display cannot be wiped with detergent or ethanol. Wipe it gently with a soft cloth. Medical air disinfection machine
During the use of the medical air disinfection machine, the maintenance of the filter is generally divided into two forms. One is cleaning. If you find too much dust on the filter, wash or replace the filter immediately with water. The second is to replace the filter, open the filter cover, remove the filter from the machine, replace the new filter, and then install the filter to the machine. UV light management: UV radiation intensity can affect the effect of air disinfection. Therefore, at least once a month, ethanol cotton balls are applied to prevent the radiation strength of UV lamps from being affected by dust. Maintenance of negative oxygen ion generator: Due to the dust reduction effect of negative oxygen ions, after long -term use, a large amount of dust will be accumulated near the air outlet of the generator, which should be removed regularly. When cleaning, cut the power first, wipe it with a soft dry cloth or a little ethanol, and not to wash it with water. The induced probe and the display cannot be wiped with detergent, ethanol, etc., and wipe it gently with a soft cloth.